SV-5Q automobile exhaustemission analyzer, used to measure the concentration of HC,CO,CO2, O2, NO
Minimum Order Quantity: 1 Set/Sets
Supply Ability: 100 Set/Sets per Month
Payment Terms: L/C,T/T,Western UnionContact Supplier
Packaging Detail: carton box
Delivery Detail: within 3 days
SV-5Q automobile exhaust emission Analyzer,
In accordance with non-light-disguising infrared absorption method. The main technical parameters
1. Environmental temperature
Temperature-5-40 humidity: 95% Air pressure:70.0-106kPa Power: AC 220V±10%;50Hz± 1Hz
2.Measuremnt scopeHC: 0-1000010-6ppmvol CO: 0-10.0 10-2(%)vol
CO2: 0-20.0 10-2 (%) volO20-25.0 10-2(%)volrotate speed0-10000rpm
NO: 0-5000 10-6ppmvoloil temperature0-120
3.Sampling methodtail gas: sampling directly .The length of sampling tube: 5m,the length of probe:900mm.
Oil temperatureinserting the head of thermo scope into the staff gauge of engine lubricant, the length of which is the same as that of oil staff gauge; blocking with rubber plug to plug up engine oil from ejecting: length of lead: 5m.
Rated rpm Non-contact measurement method. Less than 20cm to the petrol high-voltage flexible cord, measuring petrol rotate speed.
5.resolutionCO 0.01%volHC: 1ppmvolnormal hexane equivalent CO2: 0.01%volO2: 0.01%vol
NO: 1ppmvol rate speed 1rpm oil temperature 0.1
6.indicated errorCO±0.06% vol or±5% relative error. HC: ±12ppm vol or ±5% relative error
CO2: ±0.5% vol or ±5% relative error O2: 0.1% vol or±5% relative error
NO±25ppm vol or±5% relative error rate speed ±10rpm(0-10000rpm) ±1% measured value (>10000rpm)
7.Stabilityafter preheating, zero excursion and measurement distance excursion in 4 hours should not be over the allowance value of sample value error
8.Repetitiveness: ≤1/2 of absolute of the allowed sample value error
9.Output signal: Number volume: RS232 serial massage exit .